|Category:||Diagnostic Medical Device Service||Description:||Lens Repair|
|Service:||Repair Service For Probes||Warranty:||60 Days After Fixed|
|Condition:||Original||Purpose:||Diagnosis For Ultrasound|
Toshiba PVU 375BT Ultrasound Probe Repair,
Lens Replacement Ultrasound Probe Repair
Other hot Toshiba models we can repair:
|Basic Ultrasound Transducer Types|
|2||convex or curved array Transducers|
|7||Phased Array Transducers|
One may be facing several types of problems while using the ultrasound machine probes. They may be one or more of the following:
1. Poor quality image: Poor image may be due to
*Cracked or air bubbles on lens
*Tears on the sheath
2. Physical damage or broken housing: Physical damage may be due improper handling, dropping of probe, poor storage etc.
3. Noise: Noise may be due to broken cable, damaged / broken pins in connector or electrical interference.
*Breakage of probe connector pins may be caused by improper insertion or connection.
*Damage to cable occurs due to cables being ridden by some wheels or closure of drawers without proper placement of the probe inside, etc.
4. No image: If there is image being received, it could be a result of probe cable – connector breakage, probe compatibility issue or software issue.
Warnings and cautions of ultrasonic probe
★The ultrasonic probe is a valuable device. It must be careful in the process of use. Avoid dropping, impact, or abrasion to transducers.
★When installing or removing the probe, first switch off the power and then operate it carefully.
★Avoid rapid and extreme temperature changes, as well as lengthy exposure to direct sunlight or a strong ultraviolet light source.
★Do not use sharp objects to penetrate the acoustic lens.Once the acoustic lens is damaged, the coupling gel is easy to enter the interior of the probe and damage the piezoelectric element.
★Do not soak transducer in any liquid above the recommended level as stated in the user manual for your system, please operate according to the manufacturer's instructions, otherwise it will lead to circuit failure or even burn out.
★Do not disinfect at high temperature, because the probe is equipped with piezoelectric ceramics, high temperature will weaken the effect.
★Before use, carefully check whether the housing and cable are damaged, so as to prevent the probe from high voltage injury.
★After the probe is used, the residual coupling gel on the probe must be wiped clean to prevent rats or other animals from gnawing the lens.