|Category:||Diagnostic Medical Device Service||Description:||Repair Motor|
|Service:||Motor Problem||Warranty:||60 Days After Fixed|
|Condition:||Original||Purpose:||Diagnosis For Ultrasound|
RIC5-9A-RS 4D Ultrasound Probe,
Motor Repair 4D Ultrasound Probe
GE RIC5-9A-RS 4D Probe Motor Repair Motor Control Failures
Motor control failures
Other hot GE models we can repair:
Common ultrasound Probe Damage(Specialty 3D/4D Volume probes)
|Common ultrasound probe damage||Solutions|
|Lens cap damage||Lens cap replacement|
|Fluid and oil leaks||Lens cap replacement, oil reservoir repair and replacement|
|Cable cuts||Cable patches, possible cable replacement|
|Inoperative steering||Motor repair|
|Connector housing electrical damage||Minor electrical repairs, pin module replacement|
|Weak or dead elements, dropouts in an image||Array ball replacement, oil reservoir repair and replacement|
Warnings and cautions of ultrasonic probe
★The ultrasonic probe is a valuable device. It must be careful in the process of use. Avoid dropping, impact, or abrasion to transducers.
★When installing or removing the probe, first switch off the power and then operate it carefully.
★Avoid rapid and extreme temperature changes, as well as lengthy exposure to direct sunlight or a strong ultraviolet light source.
★Do not use sharp objects to penetrate the acoustic lens.Once the acoustic lens is damaged, the coupling gel is easy to enter the interior of the probe and damage the piezoelectric element.
★Do not soak transducer in any liquid above the recommended level as stated in the user manual for your system, please operate according to the manufacturer's instructions, otherwise it will lead to circuit failure or even burn out.
★Do not disinfect at high temperature, because the probe is equipped with piezoelectric ceramics, high temperature will weaken the effect.
★Before use, carefully check whether the housing and cable are damaged, so as to prevent the probe from high voltage injury.
★After the probe is used, the residual coupling gel on the probe must be wiped clean to prevent rats or other animals from gnawing the lens.